Absolute and relative reliability of isokinetic and isometric trunk strength testing using the IsoMed dynamometer. Roth R(1), Donath L(2). extension measurements using the IsoMed system, Isokinet Exerc Sci knee extension: A reproducibility study using the IsoMed dynamometer. The purpose of this study was to determine relative as well as absolute reproducibility for the IsoMed dynamometer in measuring Peak.
|Published (Last):||16 June 2006|
|PDF File Size:||9.75 Mb|
|ePub File Size:||15.41 Mb|
|Price:||Free* [*Free Regsitration Required]|
Total knee isimed TKA is a common intervention to improve quality of life after osteoarthritis. A high BFD was evident in the eccentric mode in 0200 groups highlighting a need for particular strengthening. In case of using a TFT monitor with touch-screen-function, you can work completely without mouse and keyboard. Both a desktop version see picture as a wall version, eventually with ejectable VESA arm are possible. Although, the skiing intervention affected the CSA of RF in both legs quadriceps muscle strength increased exclusively in the involved leg suggesting that most adaptations enabling strength gains were neural.
Click on the images to get more details on the right. The response is checked by repeatedly measuring the test source entered in the system which is used for calibration and comparing the mean value with the actual target value.
In conclusion, high reproducibility was shown for the IsoMed dynamometer in measuring PT during concentric and eccentric knee extension. Repeated measures analysis of variance ANOVA was applied for detection of systematic errors over sessions. Other PC systems possible on request e.
We use a holder for that, mounted between the monitor and the original monitor holder. MED offers an automated, shielded syringe filling system as an accessory to the PC-based dose calibrator. Home – Forschung – Publikationen. Moderate to high reproducibility was shown for the IsoMed system in measuring PF during concentric and eccentric isokinetic leg extension.
Systematic assessment of muscle strength of the lower extremities throughout the annual training cycle in athletes is crucial from a performance perspective for the optimization of the training process, as well as a health perspective with regard to injury prevention. Limit Results 25 50 A reproducibility study using the IsoMed dynamometer.
We recommend using a long-lived Cs test source with approx. Thirty four physically active male subjects mean age: If you need any further information on our RPM software, please call us or use our contact form! Peak Force PF of forty-one healthy male subjects mean age: The Mann-Whitney test showed no significant differences in the peak torques or ipsilateral ratios between the two groups.
Abstract Systematic assessment of muscle strength of the lower extremities throughout the annual training cycle in athletes is crucial from a performance perspective for the optimization of the training process, as well as a health perspective with regard to injury prevention. The mini-PC is available as a small desktop-PC or as a microtower and is mainly used in those locations where it is possible to build in the system, for example behind the cover of a hot cell or below a work bench.
Virtually every modern PC system can be used, regardless of manufacturer’s brand or design. Scand J Med Sci Sports.
The measuring chamber is connected via USB interface. The results of all measurements can be represented in tabular form.
Intraclass correlation coefficients ICC 2,1 were calculated to assess relative reproducibility, whereas absolute reproducibility was determined by the standard error of measurement SEM.
In addition to the optimized device software, which is also Windows 7 compatible, the measuring chamber is now connected with the PC system via USB interface. The deviation and the maximum spread between the individual measured values are displayed. Dose Calibrator Product description Versions Types of measuring chambers. Detection of systematic errors was performed using paired sample t-tests, relative and absolute reliability were examined using intraclass correlation coefficient ICC 2.
The intervention group skied three times a week whereas the control did not change their daily routines isomsd a period of 3 months. Isokinetic devices are highly rated in strength-related performance diagnosis.
However, even years after TKA, ksomed muscle strength is still impaired leading to a consistent side-to-side asymmetry in subjects with unilateral TKA Maffiuletti et al. The dose calibrator uses a gas filled chute ionization chamber as a detector system.
The test source is delivered in a lead shielding. Further nuclides can be added. Nuclide and activity of the test source, however, are freely selectable.