Hp B Pdf User Manuals. View online or download Hp B Technical Data Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 year warranty. Our staff have extensive backgrounds in T&M, totalling over OPERATING AND SERVICE MANUAL PART NUMBER Instruments Covered by this Manual. HP Model A Signal Generator with .
|Published (Last):||4 November 2013|
|PDF File Size:||3.10 Mb|
|ePub File Size:||13.28 Mb|
|Price:||Free* [*Free Regsitration Required]|
Rack Flange Kit Option 8656v takes about ms for the modulation accuracy AM or FM to be within tolerance from the time the modulation audio oscillator is initially turned on. The Data message that is sent will cause the 7 least significant HP-IB data lines to be placed in both their true and false states.
A correct decrease in the increment setting re-enables the step-up and step-down keys.
Full text of “HP: B B final”
These procedures do mamual check whether or not all of the Signal Generator program codes are being properly interpreted h executed by the Signal Generator, however, if the front-panel operation is confirmed to be working properly and its HP-IB capability operates correctly, then there is a high probability that the Signal Generator will respond properly to all of its program codes.
TTL low to recall next storage registe r contents. Connect the one-inch loop antenna to the analyzer through the amplifier as shown in Figurefor the signal path shown as 5,6. If you require a 1 or 5 MHz time base output, follow the procedure detailed below.
Absolute Level Accuracy must be tested on the Signal Generator using the frequencies fron: Each illustration is accompanied with directions and comments to provide further information. The HP Kanual is set at the factory for high impedance, refer to section 2 of the Operation and Calibration Manual if you want to set the input impedance to 50S1.
Step the frequency up 50 MHz to MHz, Exceeding this limit may cause damage to the KP SB. Section 2 Installation, provides infonnation about initial inspection, preparation for use including time base selection, and HP-IB address selection for remote operationand storage and shipment. A 10 MHz crystal reference for increased frequency accuracy and stability is installed.
Describes any miscellaneous information or information to clarify the feature or function capabilities. Directions for Recalling a Sequence 1.
Decreasing the frequency increment reactivates the key. When the Abort message is received, the Signal Generator becomes unaddressed anc stops listening.
Hewlett-Packard B Signal Generator
Refer to Figure for an example of the oscilloscope measurement display. The waveform displayed should be similar to that shown in Figure The combinatioo is made up of unique parts which include both functions.
For example, h; input in dBm generally will take about 30 ms from majual terminator to the end of software execution. You will always return to this menu when you exit the selected function. If the contents are incomplete, if there is mechanical damage or defect, or if the Signal Generator does not pass the electrical performance test notify the nearest Hewlett-Packard ofHce.
HP 8656B Manuals
Keystroke Sequence Press an internal Source key, the FM Function key, the desired Data keys, and a valid Units key you do not need to press an Internal Source key if you require a 1 kHz modulation rate.
Unless otherwise stated the information applies to all three Signal Generators. Table lists the paramaiers that have been loaded from this file. This is because when the Amplitude or Frequency is changed when Pulse Modulation is selected, the HP B performs a calibration on the output level. Oscilloscope Display NOTE If the oscilloscope does not have adequate bandwidth at the higher frequencies, proceed with the following steps using a crystal detector.
Then the amplitude of the reflected signal from the RF source is set to an equal level with the Device Under Test port shorted. HP does not warrant that the operation of the instrument, or software, or ftrmware will be uninterrupted or error free.
Replacement A4 assemblies are shipped from the factory without the attenuator pad installed. Indications The selected set-up is displayed as long as the last Data key remains pressed and will remain for 1 second after the second Data key is released.
Connect the digital multimeter to the test points shown in the table below and adjust for a 0. Example Enable a carrier frequency tuning value of 10 kHz assuming a starting value of 10 MHz.
The display will also flash after the second numeric value is entered. Data can be any number of arbitrary length in fixed point notation.