of manual HP · Calibration procedure for HP Model A. Catalog pages, covering HP A. Hp A Pdf User Manuals. View online or download Hp A Operating And Service Manual. View and Download HP A operating and service manual online. DIGITAL VOLTMETER. A Measuring Instruments pdf manual download.
|Published (Last):||25 May 2012|
|PDF File Size:||12.97 Mb|
|ePub File Size:||19.14 Mb|
|Price:||Free* [*Free Regsitration Required]|
| HP A digital voltmeter restoration
Page Service A’s Outguard can be half-split. A procedure to A30JA Clock: Make sure both instruments are being powered from same mains line socket, and very same connectors, cables are being used to reduce variables in signal path.
Shipping cost cannot be calculated. Catalog pages, covering HP A.
HP 3456A Operating And Service Manual
There is last thing left before actual processor — RAM chips. I f the V and V Ranges are noisy, set the 2. The information in the guide is most of the A’s operating characteristics, including remote Once you have decided what you want the A to do, mwnual codes. Only single units were tested, which were not had recent calibration.
Optimizing Reading Rate By simply keeping the temperature of the ‘elated. This is very easy if you use a wire-wrap socket having long pins. Slope Sequence During Rundown. This in turn brings decoder logic into question, as keyboard logic should be scanned frequently to check if any keys pressed. Page J Line A count loop is set up.
HP A OPERATING AND SERVICE MANUAL Pdf Download.
I f Test fails – 3. Make sure A2OQI03 is not open. Up to channels can be scanned per second without computer interaction. Using the sel-up information i n Figure 1 1check for the waveform shown in Ihe figure at A30U l 9 pin 3. Page If you program more than one condition in Figure 9, you will want to know which condition caused the RQS Line 2: Page Turn the A On.
Other operating characteristics are the same as the ACV function. Used lor storing the resistor value for the dBm Math operation or for recalling readings taken in the A’s Reading Storage mode. All data shown below in clickable graphs, and details available for download 34456a Excel-table. To bring the reading into tolerance.
Longwood, Florida, United States. I used soft duralumin AM2 sheets, and drawing was performed to fit chassis.
Depressing the button first places the state with the new measuring lead configuration. Storing into register is similar to the front panel method. The Converter changes the input voltage to a dc gp and then applies it to the Input c. Backdating Model A j. Set the A to the following functions and from the A30 board. Page 22 Can store up to most recent readings. He said that there is about 1uV of voltage change for 1uA of current change. Better quality copy, part 1 and part 2. El l “” J1-“”‘”‘ “‘ “.
Signal wired like shown on digital board A4 schematics marked green: Page3546a Inguard Chassis-Front and k. Learn More – opens in a new window or tab.
Service A’s Outguard can be half-split. Model A Jp Delivery in 3 days. Model A General Information Table ‘ Read more about the condition. To print the manual completely, please, download it.
There is also optional filtering circuits, which can be disabled.
The checksum for a good binary ROM image is 1C