The Pocket Colorimeter II™ has two manuals. The Basic and Full User Manuals can be found below. All Pocket Colorimeter™ II instruments. This symbol, if noted on the instrument, references the instruction manual Hach Pocket Colorimeter™ instruments* are low-cost, high-quality filter photometers. Consult HACH’s Pocket Colorimeter II brochure on ArchiExpo. Page: 1/4.
|Published (Last):||2 January 2014|
|PDF File Size:||18.89 Mb|
|ePub File Size:||7.20 Mb|
|Price:||Free* [*Free Regsitration Required]|
Got it, continue to print. Page 28 A user calibration or a user-prepared chlorine standard may be required by hacb regulatory official or agency. Place the blank in the cell The arrow should indicate holder with the diamond Samples must be Tightly cover the cell with the instrument cap.
International Customers Hach maintains a worldwide network of dealers and distributors. Page 49 Section 2 Instrument Manual 2— Page cologimeter High range free chlorine determinations are subject to variable levels of interferences from monochloramine. Reagents can be relled by contacting Hach or your Hach distributor.
DelAgua – Product – – HACH – Hach Pocket Colorimeter II, Manganese, Low Range
Operation is subject to the following two conditions: Battery Installation Instrument Operation, continued Battery Installation Figure 1 on page 2—7 provides an exploded view of battery installation. Open the catalog to page 1. Instrument Hacb, continued 2. Each instrument comes in a ready-to-use kit that includes a manual, sample cells, and a sturdy custom carrying case.
To ensure the protection provided by this equipment is not impaired, do not use or install this equipment in any manner other than that which mankal specified in this manual.
Enter the concentration value and absorbance value of the first data pair S0, A0. Insert the reagent blank or deionized water into the meter and cover with the cap. The Online Architecture and Design Exhibition.
To edit the S0 value, press. Colorimetr least two data pairs concentration and absorbance are required. Precautionary Labels Please pay particular attention to labels and tags attached to the instrument.
Hach Pocket Colorimeter™ II for Chlorine or Manganese
Error Codes Error Codes When the instrument cannot perform the function initiated by the operator, an error message will appear in the display. Forums Documents and Manuals.
My Bench Order History Sign out. Page 71 User-Entered Calibration, continued 5. Backlit Display, Data Logging, Recall Clear readings even under difficult conditions, thanks to lighting pockdt large figures on the display. Comments to this Manuals Your Name.
Repeat for all points to be deleted. User-Entered Calibration, continued Instrument Operation, continued Battery Installation Figure manuual on page 2—7 provides an exploded view of battery installation. Open the catalog to page 3. The curve can extend from 0 to 2.
Place the blank into the cell holder, with the diamond The arrow should indicate the Samples must be Note Use a TenSette pipet to add 0. User-Entered Calibration, continued 3. A flashing arrow indicates MENU the current range.
NOTE Information that requires special emphasis. User-Entered Calibration, continued 4. Many are based on EPA-approved methods. This feature allows the factory default calibration curve to be adjusted with a known standard. Colofimeter to pH 6—7 with 1 N Sodium Hydroxide. The digit to be edited will flash.