Keyboard Layout v Fluke B/1 96B-C/B-C Service Manual List of Figures For the mA. current D pin 3 is connected to pin 1, and pin 13 is. Manual de Servicio Fluke b – Ebook download as PDF File .pdf), Text File . txt) or trigger level mV) or (R the switches in D are all open. F HP Deskjet F HP Deskjet D HP Deskjet D HP Deskjet D . would work while others would not, and maybe it’s some strange fluke.
|Published (Last):||24 October 2005|
|PDF File Size:||6.74 Mb|
|ePub File Size:||19.16 Mb|
|Price:||Free* [*Free Regsitration Required]|
Full text of “FLUKE “
As a result the output voltage will become lower. Compatible with HP Envy Officejet Deskjet Printer. Ensure that the rubber sealing ring fuke 5 for the power connector is present Disassembling the Test Tool 6.
Press to select maximum brightness calibration Contrast CL Do a battery refresh: If more than one fluek groups are available, they will be separated by a vertical line. Time Base Verification 4. Remove the display from the mounting frame. Initially they were fine and I ordered a second set.
If both channels fail and Channel A is OK. Ohms Input Protection and Clamp When a voltage is applied to the input in the Ohms function V, V and V will flyke the voltage on the current source. Unpack the zip file on the usb stick. Adjust the 5 A output voltage until the displayed trace has the amplitude indicated under the appropriate column of Table This happens if only the mask software is running. Apply a kHz sine wave with a peak-to-peak value of d560 divisions to the inputs.
Press SAVE to save the actual test tool settings in setup memory 1. Continue as indicated in section 5. If VB conducts, the circuit consisting of the primary winding of transformer T and C 1, will start oscillating at its resonance frequency. Unscrew the four screws item 8. TheN ffluke is set to 6x via D With the bottom cover up, start the screws item 20 into the square nuts, then press in on the bottom holster to latch the tabs on the top case.
Tear off the plastic tape; Step 3: Probe dc calibration, can only be done if probe D15660 mutual relation should not change.
HP C9352AE (22) Multi Colour Cartridge for Inkjet Printer
If you proceeded to calibration step N for example step CLthen return to a previous step for example step CLand then calibrate this step, the complete final calibration becomes invalid; then you must repeat the calibration starting at 5. If a square wave is present on pin 49 may be not the correct value: They quickly replied and had me try a few things to remedy my problem.
Set the 5 A to source the appropriate resistance value for the first test point in Table The hinge consists of a circular raised rim in the tilt stand that is located over a circular lowering in the bottom case. Volts Peak Measurement Verification Points Adjust the tri mm er screw in the probe housing until a pure square wave is displayed.
Full text of “FLUKE B, B, C, B,C Service”
The sensitivity ranges are indicated in the left and right lower display edge in gray. Find the right part Make sure this fits your model. To display the calibration date and – number: This defines the initial test tool setup for each verification.
The latch remains attached to the connector body. There are no serviceable parts on the LCD unit. PCA Unit Assembly 7- 1. Operating Time 4 hours Charging Flue 4 hours Allowable ambient temperature during charging: To monitor and control the battery charging process, the P-ASIC senses and buffers various battery signals, as temperature, voltageand current.
The HF decade attenuator contains four separate current input amplifiers, which are connected to external capacitive dividers: PC F1560 Unit Assembly 7- 1. The VI gate is set to 3 V by N 1 54 1 output pin 1. If one or more voltages are correct, then check the rectifier diodes, coils, and capacitors of the incorrect voltage s b.
They are drawn in the circuit diagrams for PCA layout purposes. Use NO 50 Ohm termination! Disconnect the blue keypad foil item 4 flat cable, and the white LCD item 6 flex cable.