l CMW WCDMA RF Measurement l CMW LTE RF Measurement l Live Demo .. manual operation using CMW l step by step guide according to. View and Download Rohde & Schwarz CMW user manual online. Wideband Radio Communication Tester. CMW Test Equipment pdf manual download. Manuals and User Guides for Rohde & Schwarz CMW We have 1 Rohde & Schwarz CMW manual available for free PDF download: User Manual.
|Published (Last):||19 June 2016|
|PDF File Size:||1.88 Mb|
|ePub File Size:||13.44 Mb|
|Price:||Free* [*Free Regsitration Required]|
The “Startup” dialog provides the following control softkeys that you can activate while the startup is in progress. Basic Instrument Functions Use this connector to synchronize other instruments to the tester. The position of certain messages is indicated by colored bars: What does this mean in detail? As a result the dialog closes and the main view of the selected application is opened. If a bit is set, a 1 to 0 transition in the corresponding bit of the associated condition register causes a 1 to be written in the associated bit of the corresponding operqting register.
Queries the name of all network nodes detected by the subnet monitor. Negative error numbers are reserved by the SCPI standard.
Rohde & Schwarz CMW Manuals
By pressing mwnual button you iperating print only current page. Our products and the auxiliary equipment they require are designed, built and tested in accordance with the safety standards that apply in each case. Rf Path Settings generators Use the “Go to Key Facts Base model: According to the effect they have on the instrument: Parameters for setting and query: LEVel This command line is written in its full length and contains two commands separated from each other by the semicolon.
In manual control, a measurement context may consist of several views, providing different types of measurement results. These settings control the routing of input signals, the correction of the input power, the RF analyzer and trigger system.
Returns a list of paths for the STATus: Wait until the end of the operting cycle of measurement 1 and retrieve the results. A generator is in one of the following generator states: Connection States The con- nection states and state transitions are manuql refer to the description of the individual signaling application.
Returns the defined alias entries and the assigned directories.
Pdf Download | Atec Rohde-Schwarz-CMW User Manual (72 pages)
A signaling generator is in one of the following cell states: Page 20 Instrucciones de seguridad elementales Funcionamiento 1. Remote commands allow also to query information about previous corrections. This section provides basic information related to both modes. No restriction is placed upon the combination of channels: You create scripts to automate repeating settings, tests and measurements.
But they apply for example to “IQ out – RF in” scenarios.
Rohde & Schwarz CMW500 User Manual
QUEStionable register indicates whether the data currently being acquired is of questionable quality. If a bit is 1 in the enable register and its associated event bit transitions to true, a positive transition will occur in the summary bit reported to the next higher level.
Signaling generator turned off, resources either released or partially reserved from the last time the generator was turned on. OPERation register hierar- chy. The LED blinks to indicate that the reference frequency is not locked. In other words, waits until an AND operation between the bit patterns yields a positive result. Assume that a trace or a bar graph contains a series of different measurement points. Please enter the characters shown in janual image.
Comma separated list of all supported values. Macro execution must be enabled deliberately ,anual every remote connection. To print the manual completely, please, download it. They can contain measurement results, instrument settings and information on the instrument status. The measurement is stopped after the number of measurement intervals defined by the “statistic count”.
Atec Rohde-Schwarz-CMW500 User Manual – Download
Positioning The Instrument Query the measurement substates of measurement 2. The length of a single connecting cable should not exceed 2 m, if many devices are used, it should not exceed 1 operatiing.
Getting Started Cjw500 describes both the base software and the installed firmware applications. Returns the contents of the CONDition part of the status register.
Save files are stored in a default directory on drive D: Page List of all nodes detected in the subnet, including the instrument itself. If a regular expression is defined, the command is only applied to the registers matching the filter criteria.